Integrated ultrasonic imaging systems based on CMUT arrays: Recent progress
AuthorWygant, Ira O.
Yeh, David T.
Ergün, Arif Sanlı
Khuri-Yakub, Butrus Thomas
MetadataShow full item record
CitationWygant, I. O., Zhuang, X., Yeh, D. T., Nikoozadeh, A., Oralkan, O., Ergun, A. S., . . . Khuri-Yakub, B. T. (2004). Integrated ultrasonic imaging systems based on CMUT arrays: Recent progress. Paper presented at the , 1 391-394 Vol.1. doi:10.1109/ULTSYM.2004.1417745
This paper describes the development of an ultrasonic imaging system based on a two-dimensional capacitive micromachined ultrasonic transducer (CMUT) array. The transducer array and front-end electronics are designed to fit in a 5-mm endoscopic channel. A custom-designed integrated circuit, which comprises the front-end electronics, will be connected with the transducer elements via through-wafer interconnects and flip-chip bonding. FPGA-based signal-processing hardware will provide real-time three-dimensional imaging. The imaging system is being developed to demonstrate a means of integrating the front-end electronics with the transducer array and to provide a clinically useful technology. Integration of the electronics can improve signal-to-noise ratio, reduce the number of cables connecting the imaging probe to a separate processing unit, and provide a means of connecting electronics to large two-dimensional transducer arrays. This paper describes the imaging system architecture and the progress we have made on implementing each of its components: a 16×16 CMUT array, custom-designed integrated circuits, a flip-chip bonding technique, and signal-processing hardware.
SourceProceedings - IEEE Ultrasonics Symposium
Showing items related by title, author, creator and subject.
An integrated circuit with transmit beamforming and parallel receive channels for 3D ultrasound imaging: testing and characterization Wygant, Ira O.; Jamal, Nafis; Lee, Hyunjoo J.; Nikoozadeh, Amin; Zhuang, Xuefeng; Oralkan, Ömer; Ergün, Arif Sanlı; Karaman, Mustafa; Khuri-Yakub, Butrus T. (IEEE, 2007)The cost and complexity of medical ultrasound imaging systems can be reduced by integrating the transducer array with an integrated circuit (IC). By incorporating some of the system's front-end electronics into an IC, bulky ...
Güldiken, Rasim Oytun; Zahorian, Jaime S.; Gürün, Gökçe; Qureshi, Muhammad Shakeel; Balantekin, Müjdat; Tekeş, Coşkun; Hasler, Paul E.; Karaman, Mustafa; Carlier, Stephane; Değertekin, Fahrettin Levent (IEEE, 2007)This paper presents the experimental results on forward-looking Intravascular ultrasound (FL-IVUS) using dual-annular-ring CMUT arrays. The array has a diameter of 1mm including bondpads which consists of separate, concentric ...
Volumetric imaging using 2D capacitive micromachined ultrasonic transducer arrays (CMUTs): Initial results Oralkan, Ömer; Ergün, Arif Sanlı; Cheng, Chinghsiang; Johnson, Jeremy A.; Karaman, Mustafa; Lee, Thomas H.; Khuri-Yakub, Butrus Thomas (IEEE, 2002)This paper presents the first volumetric images obtained using a 2D CMUT array with through-wafer via interconnects. An 8×16-element portion of a 32×64-element array flip-chip bonded onto a glass fanout chip was used in ...