Phased subarray imaging for low-cost, wideband coherent array imaging
YazarJohnson, Jeremy A.
Ergün, Arif Sanlı
Khuri-Yakub, Butrus Thomas
Üst veriTüm öğe kaydını göster
KünyeJohnson, J. A., Oralkan, O., Ergun, A. S., Demirci, U., Karaman, M., & Khuri-Yakub, B. T. (2003). Phased subarray imaging for low-cost, wideband coherent array imaging. Paper presented at the , 2 1875-1878 Vol.2. doi:10.1109/ULTSYM.2003.1293280
The front-end hardware complexity of conventional full phased array (FPA) imaging is proportional to the number of array elements. Phased subarray (PSA) imaging has been proposed as a method of reducing the hardware complexity-and therefore system cost and size-while achieving near-FPA image quality. A new method is presented for designing the subarray-dependent interpolation filters suitable for wideband PSA imaging. The method was tested experimentally using pulse-echo data of a wire target phantom acquired using a 3.2-cm. 128-element capacitive micromachined ultrasonic transducer (CMUT) array with 85% fractional bandwidth at 3 MHz. A specific PSA configuration using seven 32-element subarrays was compared to FPA imaging, representing a 4-fold reduction in front-end hardware complexity and a 43% decrease in frame rate. For targets near the fixed transmit focal distance, the mean 6-dB lateral resolution was identical to that of FPA, the axial resolution improved by 4%, and the SNR decreased by 5 dB. Measurements were repeated for 10 different PSA configurations with subarray sizes ranging from 4 to 60. The lateral and axial resolutions did not vary significantly with subarray size; both the SNR and contrast-to-noise ratio (CNR) improved with increased subarray size.
KaynakProceedings of the IEEE Ultrasonics Symposium
Başlık, yazar, küratör ve konuya göre gösterilen ilgili öğeler.
Güldiken, Rasim Oytun; Zahorian, Jaime; Balantekin, Müjdat; Değertekin, Fahrettin Levent; Tekeş, Coşkun; Şişman, Alper; Karaman, Mustafa (IEEE, 2006)We investigate a dual-annular-ring CMUT array configuration for forward-looking intravascular ultrasound (FL-IVUS) imaging. The array consists of separate, concentric transmit and receive ring arrays built on the same ...
Şişman, Alper; Zahorian, Jaime S.; Gürün, Gökçe; Karaman, Mustafa; Balantekin, Müjdat; Değertekin, Fahrettin Levent; Hasler, Paul E. (IEEE, 2009)Side-looking (SL) IVUS probes are extensively used for management of cardiovascular diseases. Currently SL-IVUS imaging probes use either a single rotating transducer element or solid-state arrays. Probes with single ...
Damar içi öne bakan ultrasonik görüntüleme gerçek zamanlı hacimsel (üç boyutlu) görüntülemeye imkan vermesi bakımından yaygın olarak kullanılmaktadır. Damar içi görüntülemede fiziksel boyutların küçük olması nedeniyle ...