Evaluation of CMUT annular arrays for side-looking IVUS
Zahorian, Jaime S.
Değertekin, Fahrettin Levent
Hasler, Paul E.
MetadataShow full item record
CitationSisman, A., Zahorian, J., Gurun, G., Karaman, M., Balantekin, M., Degertekin, F. L., & Hasler, P. (2009). Evaluation of CMUT annular arrays for side-looking IVUS. Paper presented at the 988-991. doi:10.1109/ULTSYM.2009.5441564
Side-looking (SL) IVUS probes are extensively used for management of cardiovascular diseases. Currently SL-IVUS imaging probes use either a single rotating transducer element or solid-state arrays. Probes with single rotating piezoelectric transducer have simple front-end, but have fixed focused operation, and suffers from motion artifacts. Solid-state SL-IVUS imaging probes use piezoelectric transducer arrays and electronic beam-forming. Synthetic phased array processing of signals detected with small-sized elements in these arrays limits the SNR achievable with these probes. In this study, we explore a new SL-IVUS probe architecture employing rotating phased annular CMUT arrays. We tested and compared imaging performance of the existing and proposed probe configurations through simulated point spread functions. We also two fabricated sample annular array designs operating at 20-MHz and 50-MHz. Our experimental measurements on the 20-MHz array in oil shows 105% fractional bandwidth. The 50-MHz array with parylene coating shows approximately 40% fractional bandwidth measured in water. We also present imaging results acquired from wire-targets to test the experimental point-spread functions.
SourceProceedings - IEEE Ultrasonics Symposium
Showing items related by title, author, creator and subject.
Güldiken, Rasim Oytun; Zahorian, Jaime; Balantekin, Müjdat; Değertekin, Fahrettin Levent; Tekeş, Coşkun; Şişman, Alper; Karaman, Mustafa (IEEE, 2006)We investigate a dual-annular-ring CMUT array configuration for forward-looking intravascular ultrasound (FL-IVUS) imaging. The array consists of separate, concentric transmit and receive ring arrays built on the same ...
Damar içi öne bakan ultrasonik görüntüleme gerçek zamanlı hacimsel (üç boyutlu) görüntülemeye imkan vermesi bakımından yaygın olarak kullanılmaktadır. Damar içi görüntülemede fiziksel boyutların küçük olması nedeniyle ...
Johnson, Jeremy A.; Oralkan, Ömer; Ergün, Arif Sanlı; Demirci, Utkan; Karaman, Mustafa; Khuri-Yakub, Butrus Thomas (IEEE, 2003)The front-end hardware complexity of conventional full phased array (FPA) imaging is proportional to the number of array elements. Phased subarray (PSA) imaging has been proposed as a method of reducing the hardware ...